The Rohde & Schwarz common library offers a variety of information material.
EMC32-S: Immunity level not reached - abort EMS scan
EMC32-S: Is it possible to abort an EMS scan when the EMC32 goes below a certain immunity level, e.g. if the amplifier fails?
Yes, you can do that. In the EMS Scan Template, on the Leveling Mode t...
EMC32-S: Pulse train support in EMC32
I am using EMC32. How can I configure a test template with the SMB100A (including SMB-K27) or the SMF100A (including SMF-K27) to generate pulse trains for my EMC32 application?
The pulse train software option i...
FSx doesn't recognize active probe RT-ZS30
We have an issue with our FSx and the RT-ZS30 probe: the FSx does not automatically recognize the probe Could you please help us to solve this blocking issue.
You can work with the RT-ZS30 connected ...
Web interface allows several users to control the HMP
The HMP is used by two persons at the same time. One is using channel 1 and the other is using channel 2. Are they able to remotely control their channels at the same time via the web interface?
How to connect HMExplorer to the instrument
My HMExplorer does not connect to the instrument
Error when adding a device. The interface could not be opened.
The latest version of the HMExplorer should be located at h...
Changing virtual COM port to use field probe with EMC32
We are using EMC32 software for EMS measurements. Currently we have difficulties integrating a field probe. We cannot change the COM port of the field probe's USB/serial adapter and therefore cannot use the field probe w...
EMC32-S: Amplifier properties, reverse coupling path attenuation
In my hardware setup, I have two power meters connected to the amplifier, one for the amplifier FWD path and one for the amplifier REV path. When I look at the amplifier properties, I can only define the attenua...
Improve the calibration procedure for different setups
I have to perform a measurement with different settings (power, bandwidth, ...). If I use the microwave prober with an on-chip calibration substrate to perform a TOSM calibration, it would take a very long time becau...
Forward and reverse power measurements with NAP generic power meter
In EMC32, there is a generic driver for the old Rohde & Schwarz NAP power meter I would like to use this to measure forward and reverse power with EMC32-S. However, for some reason the EMC32 software does no...
Operating with two displays in parallel
For some measurement tasks, it is very convenient to have the measurement on the ZVL display and also on an external display. Requirement: The ZVL needs a DVI connector on the back. This FAQ explains the setup starting in spectrum analy...